We have created a awesome theme
Far far away,behind the word mountains, far from the countries
|Press Emissivity key (3), then press Up key (4) or Down key (2) to set the emissivity, then press Mode key (5) to confirm it. The emissivity can be changed from 0.10 (10E) to 1 (100E).|
|Here will show the emissivity data. (The default emissivity is 0.95.)|
|Press Mode key (5) for the Maximum (MAX), Minimum (MIN), Difference between MAX and MIN (DIF) and Average (AVG) modes. During the measurement, the special modes reading will be displayed beside the mode icon.|
|Press Up key (4) or Down key (2) key to change the High Alarm (HAL) or Lo Alarm (LAL), then press Meas. key (8) to confirm it. When the reading is outside the High Alarm (HAL) or Lo Alarm (LAL) limit. The High or Low icon will flash and you will hear a beep sound.|
Connect the thermocouple with Thermocouple socket (7) and put the probe in/on the target, the thermometer will display the temperature automatically without pressing any button. To see the minimum or maximum data during the probe measurement, please hold down the Up key (4) or Down key (2).
After measure high temp, the probe may remain HOT for a while.
|MAX, MIN, DIF, AVG Mode:||
Press Up key (4) for LOCK mode ON/OFF. The lock mode is particularly useful for continuous monitoring of for up to 60 minutes.
Press Down key (2) for oC or oF transferred.
|In all modes: First hold on the Meas. key (8)||and press Lock key (4) for backlight function ON/OFF.|
|and press C/ ºF key (3) for laser function ON/OFF.|
|‘Hi’ or ’Lo’ is displayed when the temperature being measured is outside of the settings of HAL and LAL.|
|‘Er2’ is displayed when the thermometer is exposed to rapid changes in the ambient temperature.|
|‘Er3’ is displayed when the ambient temperature exceeds 0C (32°F) or +50C (122°F). The thermometer should be allowed plenty of time (minimum 30 minutes) to stabilize to the working/room temperature.|
|Error 5~9, for all other error messages it is necessary to reset the thermometer. To reset it, turn the instrument off, remove the battery and wait for a minimum of one minute, reinsert the battery and turn on. If the error message remains, please contact the Service Department for further assistance.|
|‘Battery OK’: measurements are possible||‘Battery Low’: battery needs to be replaced, measurements are still possible||‘Battery Exhausted’: measurements are not possible|
|Item||Non-contact Infrared Scan function||
Thermocouple Probe Scan function
(K type; a bead wire included.)
|Measurement Range||-35 ~ +1500℃||
K Type: -64 ~ +1400C
Bead: -50 ~ +250C
|Operating Range||0 ~ 50℃|
|Accuracy||±2% of reading or 2C ±2%||±1% of reading or 1C|
|Emissivity Range||0.1 ~ 1.0 (Default 0.95)|
|Response Time||1 sec|
|Memory||Detailed 24 Memories with Temperature & Emissivity|
|Battery Life||min 14 hours continuous use (Alkaline, without Laser and Back Light.)|
|Dimensions||204.14 x 179.19 x 58.40mm|
|Weight||355g including batteries (AAA*2pcs)|